Introduce cranial drilling and attention
Product Model: ND-4011
Frequency: 0-920rpm / Hz
Stall torque: 8.5N.m
5 minutes warming: 15°C
The product can be quickly and easily loaded and unloaded, the use of quick-change parts structure, removal and cleaning more convenient and reduce operating time.
What you should pay attention to is:
- Check if the drill can be operated directly before surgery. If there is any deviation, the drill should be replaced. Check if the drill bit is sharp, if not, polish it. The drill hole should be firmly fixed, the hand cannot be shaken by conventional surgery, and the broken drill bit can be replaced immediately.
- The drill bit should be fixed firmly. A special chuck key can help lock the bit in place to avoid accidents. To avoid rust, drill saws and accessories should be placed in a dry environment.
- Make sure that the saw blade is installed correctly, and there must be no cracks, unevenness, etc.
- The battery cannot be autoclaved and is stored in the main unit for a long time. Please remove the battery promptly after surgery. If the power is off during the operation because the main unit is not running, replace the backup battery immediately.
- If the abnormal sound occurs during use, stop work immediately and contact the manufacturer or dealer to return the equipment to the factory for repair.
- This product is a professional medical device. The user must have a certain technology or a properly trained medical staff.
Wuhu Ruijin Medical Instrument Co., Ltd. is a large-scale, fully-productive electric medical tool manufacturer with more than ten years of experience.